Cellex qSARS-Cov-2 IgG/IgM Control
Cellex qSARS-Cov-2 IgG/IgM Positive and Negative Control should be tested to ensure the proper performance of the assay, particularly under the following circumstances:
2)A new lot of test kits is used;
3)A new shipment of kits is used;
4)A new test environment is used (e.g., natural light vs artificial light);
5)The temperature used during storage of the kit falls outside of 2-30 ℃;
6)To verify a higher than expected frequency of positive or negative results or negative results;
7)To investigate the cause of repeated invalid results; or
8)A new test environment is used (e.g., natural light vs. artificial light)
Control Storage Condition:
2-8 DEGREE C.
Instruction for Use:
It is recommended that the qSARS-Cov-2 IgG/IgM Positive and Negative Control be spun down at 500-1000 RPM for 30 seconds before first use. When performed properly, in addition to the presence of C Line, no line should be visible for the negative control and the G Line or M Line or both lines is/are visible for the positive controls. The positive control may contain IgG or IgM or both antibodies. Additional controls may be qualified and tested by the user.